How do I calibrate the Q-Power or Q-Gas?
Calibration applies only to Q-Gas with firmware 3.30 or newer, set for silver spot gas meter and Q-Power, set for mechanical meter. To calibrate these devices press five times the button at the bottom of the device
Note: Q-Power’s calibration will fail if there is very low electricity consumption.
If your Q-Gas has firmware version 3.29 or older, check if the sensor head is mounted correctly, take out the batteries, wait for 1-2 minutes for the device to discharge and then put the batteries back it. The sensor head must be attached both to the gas meter and the reader while you do this. It will take a couple of minutes until it adjusts. Until then it might not record any consumption.